All Notes (14)
Recovered by Brian Buckingham, Yachen Li, Emily Busch aboard NJDEP vessel with Capt. Rodney Sloan out of Atlantic City, NJ.
Comparison CTD cast done with RBR-Concerto3.
Glider recovered at same time as RU23 deployment.
glider aborted for no heap
added line in 100_nw.mi: b_arg: reqd_spare_heap(bytes) 30000
All instruments to every 4th yo
10/4 00 gmt interaction
10/3 7:00 interacted with
All sensors on all the time starting last night for the storm
alongshore north waypoint to preempt the storm
Phone not holding connection well it seems. At least not recently
Operating System: 8.3
Wetlabs ECO-Puck FLBBCD-SLC
Aanderaa Optode 5014W
In-house check 9/2/2022
Strap-on, no data feed through glider
Duty cycling all sensors, every 4th yo since deployment.
Station kept waiting for decision on staying with ru30, decided to send it onward.
Will switch to every 3rd once in deeper water.
Deployed by Nicole Waite, Brian Buckingham, MOO students Rhyan G, Jake Z, Bre D aboard R/V Rutgers with Captain Chip Haldeman.
Deployed with RU30.
Comparison CTD cast done with SB19 1581 and ballast tank computer.
Seabird CTD - s/n 0103p, Cal Date 3/27/2022
Wetlabd FLBBCDSLC - s/n 3326, Cal Date 5/4/2021
Aanderaa Optode 5014W - s/n 1504, Cal Date 4/27/2021, In-House check 9/2/2022
VMT standalone, ID59965
Setting SENSOR u_abort_min_burn_time(sec) value 600.000000
Setting SENSOR u_abort_max_burn_time(sec) value 28800.000000
Setting SENSOR u_reqd_depth_at_surface(m) value 2.000000
Setting SENSOR u_hd_fin_ap_run_time(secs) value 12.000000
Setting SENSOR u_hd_fin_ap_gain(1/rad) value 1.500000
Setting SENSOR u_hd_fin_ap_igain(1/rad-sec) value 0.015000
Setting SENSOR u_hd_fin_ap_dgain(sec/rad) value 6.000000
Setting SENSOR u_hd_fin_ap_hardover_holdoff(sec) value 30.000000
Setting SENSOR u_pitch_ap_gain(1/rad) value -1.000000
Setting SENSOR u_pitch_ap_dgain(s/rad) value -1.000000
Setting SENSOR u_pitch_ap_deadband(rad) value 0.079000
Setting SENSOR f_battpos_db_frac_dz(nodim) value 1.000000
Setting SENSOR u_min_pitch_ap_period(sec) value 2.000000
Setting SENSOR u_max_pitch_ap_period(sec) value 60.000000
Strange segment resulted in attitude_rev abort
On a climb at 6 m glider stopped ascending and pitched to level from +26. Wasn't able to climb for 30+ seconds and after that compass stopped reporting.
Downloaded MBD and MLG from event.
All sensors every +4 nth yo to sample