All Notes (15)
Optode SN 971 had some noise in readings found by Laura P. Chip and Dave checked optode in lab 1/14/2019 and found the foil was *slightly shifted and could see green light behind one side of the shifted foil. Applying pressure to the foil caused the numbers to fluctuate to a rough maximum of +- 10 micro moles. Usually < +- 5 micro moles. Optode should go back for check up and possibly new foil before being used again. Optode 3835 rated to 300 m
CC is off
SVS serial number is 121
Schuyler reported a red tide going on in the water at recovery
Recovered off Atlantic City by Schuyler Nardelli and the NJ DEP.
CTD cast with SBE19 s/n 1383, and a shall_nw.mi mission (log file 03970000)
CC is off
Beh aborts: think science crashing and not coming up in time to open log files.
Longer deployment here will be better for weather and travel, optics to every other yo, see if that helps. LISST to every 16th yo
Spent last night near Orsted buoy F230 with wave sensor ON, moving onward today
Make that every 6th
Lisst every 4th yo, wave off after storm.
Wave SVS on overnight
# ru28 Custom Config
# 8/14/2014 initial based on 2014 NJDEP 1
# 8/15/2015 disabled the filter; I don't think it does anything useful
# 2018/9/20 added lengthened abort timers
# Low density abort stuff
sensor: u_abort_min_burn_time(sec) 600 # Never drop the weight before this time
sensor: u_abort_max_burn_time(sec) 28800 # 8 hrs, at least a tidal cycle, Always drop the weight after this time
sensor: u_alt_reduced_usage_mode(bool) 0
sensor: u_alt_reqd_good_in_a_row(nodim) 1
sensor: u_alt_filter_enabled(bool) 0
sensor: u_min_altimeter(m) 2.0
sensor: u_max_altimeter(m) 8.5
sensor: u_alt_min_depth(m) 1
# Avoid old bug
sensor: m_altitude_rate(m/s) -1
sensor: u_hd_fin_ap_gain(1/rad) 1.50
sensor: u_hd_fin_ap_igain(1/rad-sec) 0.01
sensor: u_hd_fin_ap_dgain(sec/rad) 5.00
sensor: u_heading_deadband(rad) 0.087
sensor: u_heading_rate_deadband(rad/s) 0.0087
sensor: u_hd_fin_ap_deadband_reset(bool) 0 # true - reset integrator, false hold integrator
sensor: u_hd_fin_ap_hardover_holdoff(sec) 60.0
Deployed out of Shark River by DEP Rand Haldeman Waite and Evan Ramsay. Deployed 3 miles due east of inlet due to weather.
OPTODE 3835 SN 971 (previous optode cable failed, noticed in lab)
LISST-200X SN 7003 FIRMWARE 1.415
SVS603 SN and firmware to be checked!
VMT SN 1144951