All Notes (6)
ru05 with SB 0080 configured for CONVERGE/LTER 2014/2015 had the wrong calibration coefficients in the autoexec.mi file. See note below:
# 2015/09/10 firstname.lastname@example.org SB0080 was placed in glider for LTER(CONVERGE) 2014/2015 season. This payload has pucks BBFL2-337 and BB3-335
# The calibration coeffcients in the autoexec during those missions was for pucks BB3SLO-648 and BB2FLS-666
# 748, and 666 were associated with SB 0175 for California UCSB
# Changed coeffcients in autoexec.mi to the correct pucks, BBFL2-337 and BB3-335
# Changed ballast pump deadz width from 30 to 15 cc
Recovered by Megan Cimino 2/8 12 EST from Palmer station
I actually brought the pump throw back to 180 cc the other day during the high winds. I thought it was the glider but it was the winds.
Slowing back down to + 145 cc climbs to see if we can squeeze another day or 2 out of the mission.
+130 cc climbs, might be aggressive, but trying it.
Changed climb to 180 cc to save power.
Optics on every downcast.
Maintaining original ud_134 line.