All Notes (7)
Optode 972 sent to Aanderaa for re-cal / post cruise check
bpump is overheated. 16 is now in drift mode; needs to be recovered.
~17:45 EDT RU16 surfaced for device error, bpump out of service. Put bpump back into service, waiting for callback.
GliderDos A 16 >use + buoyancy_pump
BUOYANCY_PUMP:Putting device BACK INTO SERVICE: BUOYANCY_PUMP
Exiting all devices ...
1118 2 disabling Iridium cons
Tx\'ed .dbd; all looked well. Immediately sent on 3 hour missions. Dive-to changed to 98 m @ 4:47 EDT surfacing.
Flaky altimeter performance on this glider.
Replaced stock altimeter with ru27\'s.
Determined RU16\'s altimeter is still not working properly. On 4/30/10 entire glider was sent to Webb to see if they could fix the altimeter. Altimeter for this glider has never worked properly.